Michael wrote:I have ensured both my OBi200 and OBi1062 devices were hard reset before I followed your instructions.
Did you re-configure them with the correct UK settings?
If not, update both devices with the correct UK settings using the files in the links above. Doing so will also apply the correct UK digitmap(s) to both of your devices.
This can be a pain in the backside when using the OBiTALK portal, Via OBiTALK - OBi200:
OBi Expert Configuration > Manage Expert Configuration- Saved Configuration File:
Choose file and click on the "Import Config to OBi Expert
Importing a Configuration: At the bottom of the page there is an Import Configuration section, where there resides a function which allows the user to import (or upload) a previously saved XML configuration file as their OBi Expert database configuration.
When this option is selected, the OBi Expert database will import these settings and save them as the device's OBi Expert Configuration settings. Once this configuration is in the database, the user can uncheck the OBiTALK setting box to selectively invoke parameter configurations contained in the imported config file. The OBiTALK box is checked by default because the OBi Expert mode requires that the user understand the changes they make while in the OBi Expert Configuration menus can possibly alter or break functionality associated with their OBiTALK configuration. Also since OBiTALK controlled parameters change overtime, it is safer for the OBi Expert user to have to uncheck the OBiTALK Settings box in that a configuration that they previously knew to not be in conflict with OBiTALK may be in conflict the next time they go to import a config file.
Inside the OBi Expert Configuration pages, you will notice its resemblance to the OBi device web page menu, look and layout. To the right of the configurable parameter value field, there may be a red exclamation mark (!). When this mark is present, it means the parameter value next to where it appears has an OBi Expert configuration database entry that is different than the OBiTALK database entry. If the Device Default and OBiTALK boxes are left unchecked, then the OBi will use this setting in its active configuration.
On the right side of the OBiTALK Settings check box there may be an asterisk (*) indicating this parameter is currently being used in the OBiTALK portal for feature configuration. Therefore, if the symbol resides next to a parameter setting that you choose to over-ride with an OBi Expert Configuration setting, the functionality configured by the OBiTALK portal is subject to conflicted operation. It is important to note that the presence of the * may appear suddenly, when functionality is added to the OBiTALK portal that requires management of the parameter next to which the * appears. We suggest that users uncheck the OBiTALK box for those parameters NOT marked with the *, when the particular parameter value is different for a configuration outside the realm of what is available in OBiTALK. Unchecking the OBiTALK box will ensure the custom value will not be overwritten by a future OBiTALK enhancement or modification.
Remember: If you want to configure a value for a parameter as your OBi Expert configuration setting, you must have both the "OBiTALK Settings" and "Device Default" check boxes UNCHECKED. Only then, can you enter a new value to be submitted as an OBi Expert Configuration setting.
Finally, before jumping to a new page to check on settings or make modifications, please do not forget to select the "Submit" button for each page on which you have made a modification that you want to save to the device's OBi Expert configuration file.